Launched in April 1999, Landsat-7 ETM+ continues to acquire data globally. The Scan Line Corrector in failure in 2003 has affected ground coverage and the recent switch to Bumper Mode operations in April 2007 has degraded the internal geometric accuracy of the data, but the radiometry has been unaffected. The best of the three on-board calibrators for the reflective bands, the Full Aperture Solar Calibrator, has indicated slow changes in the ETM+, but this is believed to be due to contamination on the panel rather then instrument degradation. The Internal Calibrator lamp 2, though it has not been used regularly throughout the whole mission, indicates smaller changes than the FASC since 2003. The changes indicated by lamp 2 are only statistically significant in band 1, circa 0.3% per year, and may be lamp as opposed to instrument degradations. Regular observations of desert targets in the Saharan and Arabian deserts indicate the no change in the ETM+ reflective band response, though the uncertainty is larger and does not preclude the small changes indicated by lamp 2. The thermal band continues to be stable and well-calibrated since an offset error was corrected in late-2000. Launched in 1984, Landsat-5 TM also continues to acquire global data; though without the benefit of an on-board recorder, data can only be acquired where a ground station is within range. Historically, the calibration of the TM reflective bands has used an onboard calibration system with multiple lamps. The calibration procedure for the TM reflective bands was updated in 2003 based on the best estimate at the time, using only one of the three lamps and a cross-calibration with Landsat-7 ETM+. Since then, the Saharan desert sites have been used to validate this calibration model. Problems were found with the lamp based model of up to 13% in band 1. Using the Saharan data, a new model was developed and implemented in the US processing system in April 2007. The TM thermal band was found to have a calibration offset error of 0.092 W/m 2 sr ??m (0.68K at 300K) based on vicarious calibration data between 1999 and 2006. The offset error was corrected in the US processing system on April 2007 for all data acquired since April 1999.
Additional Publication Details
Radiometric calibration status of Landsat-7 and Landsat-5
Larger Work Title:
Proceedings of SPIE - The International Society for Optical Engineering
Sensors, Systems, and Next-Generation Satellites XI