The Level TROLL 100 manufactured by In-Situ Inc. was evaluated by the U.S. Geological Survey (USGS) Hydrologic Instrumentation Facility (HIF) for conformance to the manufacturer’s accuracy specifications for measuring pressure throughout the device’s operating temperature range. The Level TROLL 100 is a submersible, sealed, water-level sensing device with an operating pressure range equivalent to 0 to 30 feet of water over a temperature range of −20 to 50 degrees Celsius (°C). The device met the manufacturer’s stated accuracy specifications for pressure within its temperature-compensated operating range of 0 to 50 °C. The device’s accuracy specifications did not meet established USGS requirements for primary water-stage sensors used in the operation of streamgages, but the Level TROLL 100 may be suitable for other hydrologic data-collection applications. As a note, the Level TROLL 100 is not designed to meet USGS accuracy requirements. Manufacturer accuracy specifications were evaluated, and the procedures followed and the results obtained are described in this report. USGS accuracy requirements are routinely examined and reported when instruments are evaluated at the HIF.
Additional publication details
USGS Numbered Series
Laboratory evaluation of the Level TROLL 100 manufactured by In-Situ Inc.: results of pressure and temperature tests