A comparison of data (specific conductance, dissolved-oxygen concentration, temperature, and pH) collected by the U.S. Geological Survey flowthrough monitor, the U.S. Geological Survey minimonitor, and a self-contained commercial 'packaged-sensor' system indicates that the data obtained by means of the most complete of the three systems.
The U.S. Geological Survey flowthrough monitor is powered by 120-volt alternating current and in a heated weather-proof shelter. A pumping system brings water from the stream to sensors clustered in a sample clustered in a sample chamber located in the shelter. This instrument measures output from the senors; data are recorded in binary-coded decimal form on a 16-channel punched-paper tape recorder tape recorder housed in the shelter.
The U.S. Geological Survey's minimonitor is powered by an external battery and is housed in a weatherproof shelter. This instrument measures output of instream sensors with extension cables having underwater connectors; data are recorded in binary-coded decimal form on a 16-channel punched-paper tape recorder housed in the shelter.
The packaged-sensor system also measures output of senors housed in a packages that is submerged in the stream. It has internal power supply, no moving parts, and does not requires a weatherproof shelter; data are stored are stored in solid-state memory.
Minimonitors were installed at four sites in Ohio where U.S. Geological survey flowthrough were in operation. Two package-sensor systems also were assigned to each site and were alternated every two weeks. Detailed records were kept of (1) field measurements, for comparison with monitor-system data from each instrument, and (2) equipment problems that resulted in loss of data. Results of the comparisons shows that the flow-through monitor gave the most accurate and the most complete data.
Additional Publication Details
USGS Numbered Series
Comparison of accuracy and completeness of data obtained from three types of automatic water-quality monitors
Water-Resources Investigations Report
Dept. of the Interior, U.S. Geological Survey ;
Books and Open-File Reports [distributor],