The Storm 3 is a browser-based data logger manufactured by WaterLOG/Xylem Incorporated that operates over a temperature range of −40 to 60 degrees Celsius (°C). A Storm logger with no built-in telemetry (Storm3-00) and a logger with built-in cellular modem (Storm3-03) were evaluated by the U.S. Geological Survey (USGS) Hydrologic Instrumentation Facility (HIF) for conformance to the manufacturer’s specifications with bench tests, for recording data over the device’s operating temperature range with temperature chamber tests, and for field performance with an outdoor deployment test.
The procedures followed and the results obtained from the testing are described in this publication. The device met most of the manufacturer’s stated specifications. An exception was power consumption, which was about 10 percent above the manufacturer’s specifications. It was also observed that enabling WiFi doubles the Storm 3’s power consumption. In addition, several logging errors were made by two units during deployment testing, but it could not be determined whether these errors were the fault of the Storm or of an attached sensor.
Kunkle, G.A., 2016, Evaluation of the Storm 3 data logger manufactured by Waterlog/Xylem Incorporated—Results of Bench, Temperature, and Field Deployment Testing: U.S. Geological Survey Open-File Report 2016–1054, 9 p., http://dx.doi.org/10.3133/ofr20161054.
ISSN: 2331-1258 (online)
Table of Contents
- Description of the Storm 3 Data Logger
- References Cited
|Publication Subtype||USGS Numbered Series|
|Title||Evaluation of the Storm 3 data logger manufactured by WaterLOG/Xylem Incorporated—Results of bench, temperature, and field deployment testing|
|Series title||Open-File Report|
|Publisher||U.S. Geological Survey|
|Publisher location||Reston, VA|
|Contributing office(s)||Office of Surface Water|
|Description||iii, 9 p.|
|Online Only (Y/N)||Y|
|Additional Online Files (Y/N)||N|
|Google Analytic Metrics||Metrics page|